» Faq I want to return something – what do I do.
Goods can only be returned after Dyna’s prior written permission within 14 days after delivery, in their original, undamaged and unopened packaging and accompanied by the original relevant invoice or packing list enclosed with the shipment.
Use solid packaging to return articles and send them with sufficient postage to: Dyna Dental Engineering b.v., Vang 9, 4661 TX Halsteren, The Netherlands In case of large volume you can get in touch with +31 (0)164 271080 or [email protected] The Dyna hex driver does not fit in the hole of the screw retained crown/bridge.
Please check if you have the latest, smaller, hex driver.
This has been adjusted according to the width of the fixation screw.
If the screw fits the hole of the crown, the hex driver should fit as well.
The latch head part of the octa driver does not fit the contra angle anymore or got stuck in there.
Mechanical damage of latch-head instruments can occur above a torque of 50Ncm.
Then use a T.
instrument/driver in combination with the torque wrench in combination with the rotor bit 4x4mm square.
How do I remove a broken fixation screw.
Different options are available .
Check the instructions in the separate enclosure.
Method 1 is to place an instrument with a very narrow point between the broken screwpart and the internal thread of the implant.
Try to move the instrument’s point anti-clockwise and see if the screw loosens and can be removed.
Which side of the magnet needs to be where.
The logo on the magnet needs to be in the acrylic, so won’t be visible after finishing the denture.
Please take care not to damage the magnet.
Most steel drills are attracted by magnets, so use only drills that are not susceptible to magnetism when finishing and polishing the denture.
Couldn’t find the answer.
Or look for more information in the downloads section Downloads.